How to Maintain Hot and Cold Impact Test Chambers?
When it comes to the right way to maintain hot and cold impact test chambers, it's not comprehensive enough,so we collated and collected a number of methods and experience for Maintenance Personnel Reference.It teaches you that proper maintenance of the hot and cold impact test chambers,which can effectively extend the service life of the hot and cold impact test chambers,reduce the failure rate of the hot and cold impact test chambers, thus reducing the time and money wasted in maintenance and repair Create more economic benefits for the company.
Matters Needing Attention of the hot and cold impact test chambers:
1.The hot and cold impact test chambers should be fixed to clean the condenser every 3 months: for the Refrigeration System Using Air Cooling, the condenser fan should be repaired regularly and the condenser should be decontaminated to ensure its good ventilation and heat transfer performance;For the cooling system using water cooling, in addition to ensure its water pressure, water temperature within the specified range, but also must ensure the corresponding flow, and regular cleaning of the condenser to remove scale, in order to obtain its continuous heat transfer performance.
2.The hot and cold impact test chambers should be regularly cleaned evaporator: due to the sample of different levels of clean, in the forced air circulation, the evaporator will condense a lot of dust and other small particles, should be regularly cleaned.
3.If the water is not smooth, humidifier scaling easy to cause the humidifier dry burning, may damage humidifier, so must be regularly on the water, humidifier cleaning.
4.If the high and low temperature test box for a long time to do low temperature, when the completion of a cycle, should set the temperature of 110 degrees, a small range of the door to do two hours defrosting treatment. At the same time should adhere to each test, set the temperature near the ambient temperature, work for about 30 minutes, and then cut off the power, and clean the walls of the studio.
5.If the hot and cold impact test chambers is not in use for a long period of time, it should be regularly powered on every half-month, power-on time is not less than 1 hour, and check whether the relevant parts and components of the equipment are running normally.