This machine can be used for various kinds of connector plugging, tension life, tensile force failure, pressure life and destruction, shrapnel forward force and so on. Through the analysis of industrial computer, the load, stroke and relative strain curve of the object to be measured can be accurately measured, and the stroke, speed, target number and tentative time of the plugging force of the connector can be determined.
1.Turn on the power switch of the computer power supply → the right side of the equipment → the ship type power switch waits for the computer to enter the operating system normally.
2.Double-click the desktop test software icon → pop up the load protection screen as shown below (the protection is greater than the product test load but not greater than the full range of the sensor) Click OK to enter the settings screen.
Test Name: Shows the current file
Direction of motion: Compression tensile test
Type of test:
Silicone rubber test
Range of load measurement: reaches the power station stop or return
Range of travel determination: to reach the stop or return of the travel machine
Measuring Position Checking: Starting Measuring Position (Starting Machine Starting Counting Travel, Plug Test)
Load zero point position (calculate stroke, forward force test when load is available)
Speed of test: Speed of movement when testing machine (range:5-300mm/part, GB 25.4)
Maximum Load Value (go): insert the most vigorously, can be set according to the product specifications up and down specifications (can automatically determine OK\\NG)
Maximum Load Value (Back): Pull out the most powerful, according to the product specifications set up and down specifications (can automatically determine OK\\NG)
Load Sensor: Selected by Device Hardware Sensor
Unit: Select the appropriate unit of strength according to demand
Probe test: set the test speed: set the pressing position: set the pressing position: set the pressing position: set the pressing position: number of test times: upper and lower limit of shrapnel: save the file: enter the execution screen: click on the file: pre-set value storage: select the storage path and test name: enter the execution screen: align the pressure head to the product zero position: click on the origin to confirm that the machine should wait for the measuring point to start after rising Button test.
1.Go one-way test: the device moves the beam up to reach the stroke or load back to the original position, record the compression curve and test value. for forward force, depin test, general pressure test.
2.Go back to the bidirectional test: the device moves the beam up to the stroke or the load back to the original position, recording the compression tensile curve and the test value. for forward force, plugging force, general pressure and life test.
1.Lashing test: the device moves the beam down to pull the product to reach the stroke or load stop. Record drop-down curves and test values. for pulling pin, wire breaking failure test, general hold force test.
2.Constant pressure test: after the equipment is compressed up to the preset load, the load remains constant until the end of the time and returns to its original position, recording the compression curve and the test value.
3.Constant tension test: the equipment is stretched down to the preset load after the constant load is maintained until the end of the time to return to the original position, record the stretching curve and test value.
4.Pressure fracture test: the equipment moves the beam upward compression product to reach the stroke or load stop. Record compression curves and test values. for back PIN, plate compression failure test, general compression test.
5.Pull-off return: the device moves the beam down to pull the product to reach the trip or load the device back to its original position. Record drop-down curves and test values. for pulling pin, wire breaking failure test, general hold force test.
6.Step Load: The device moves the beam up to set the 1-stage load and then continues to compress to 2-section and so on. Record compression curve and test value for compression withstand voltage test.
7.Starting Force Pressure Test: The device moves the beam up to compress the product to reach the stroke or the load to return to the starting force position and continue the test at the starting force position next time. Record compression curves and test values. for pressure fatigue life test.
If you have any interest of the products above, or if you need any professional advice of those machines, welcome to contact us.