What Is The Reason For Non Spray Of Salt Spray Test Chamber
Salt spray test chamber is a test for corrosion of automobile parts such as hardware and metal. It can quickly detect the corrosion degree of products in temperature and humidity environment, and it can effectively improve production efficiency.
So what is the reason for such a salt spray test chamber not to spray?
1. It may be that the spray tower is blocked.
2. The water pipe is blocked and the water flow does not go in.
3, Air compressor stop operation, the air compressor button to open;
4, The general switch of air compressor outlet is not opened, and the general switch of air compressor is opened.
5, The malfunction of the solenoid valve, the failure of the pressure gauge or the pressure is too low, it is necessary to notify the maintenance company to inspect the door.
6. When the nozzle is blocked, it is necessary to carefully remove the nozzle for cleaning.
7. If the spray pressure is normal, the position of the nozzle glass is correct, but what is the reason for not spray? In this case, it is necessary to carefully observe whether the contact surface of the nozzle outgassing is fouling, if some, the dirt is cleaned up, and the spray can be carried out normally.